High quality radar level sensor provider? So what are the installation technical requirements for radar water level meters? The installation of the radar water level meter must be vertical to the object to be measured; there should be no obstructions between the measured object and the radar water level meter, otherwise it will affect the reflection of radar waves, that is, affect the measurement accuracy; the center of the radar water level meter is far from the shore of the water body. The distance must be greater than the radius of the transmitting beam, otherwise the measurement accuracy will also be affected; the installation cantilever bracket of the radar water level meter must be firm, and cannot be shaken up and down; in order to protect the radar water level meter, the water level meter can be installed on the cantilever bracket. Iron box, put the radar water level meter probe. Discover more information at radar level transmitter.
For the continuous measurement of corrosive, high viscosity, flammable and toxic liquids, liquid-liquid interface and solid-liquid interface that are difficult to measure by ordinary liquid level meter , ultrasonic level meter should be selected, but should not be used in occasions with large liquid level fluctuations . Ultrasonic level meter are suitable for measuring media that can fully reflect and transmit sound waves, but should not be used in vacuum. The ultrasonic flow meter should not be used in process equipment where internal obstacles affect the propagation of sound waves.
In addition, some silos in cement plants are very high, such as homogenizing silos of 50cm. It takes time and energy to board high silos to debug radar, so it is recommended to choose HART handheld operators that can be debugged remotely in the central control room. In the central control room, the range and other basic parameters can be set, and the radar echo waveform can be observed, and the waveform can be used for remote diagnosis and debugging, greatly reducing the on-site work intensity of the staff, to avoid the risk of climbing operation. The smart radar level gauge commonly used at present also has a function similar to “driving recorder”, that is, when the material surface mutation occurs on the scene, it can capture the radar echo waveform at that time, which is very useful for debugging the silo under complex conditions.
If the radar level meter is unreasonably selected, the interference echo cannot be handled well, and the reliability of the instrument will be reduced. Therefore, the following factors should be considered when selecting a radar level meter: Conductivity and dielectric constant of the measured medium. The measured medium is a conductive liquid or a liquid with a dielectric constant above 4. Generally, a common radar is selected. Liquids with small dielectric constants (dielectric constants below 2) and some conductive solids often use precision radars or guided wave radars due to the large amount of interference echoes.
For radar level gauges, there are many reasons for interference and many sources of interference. We analyze from four aspects: internal, external, AC and DC. Celestial and celestial interference, first of all, what is celestial interference? Celestial objects refer to the sun or other stars, therefore, celestial interference refers to the interference of their electromagnetic waves on the radar level gauge. We are very unfamiliar with Tiandian. The so-called Tiandian is usually understood as the interference of the signal of the magnetrol radar level gauge caused by the ionization of the atmosphere, lightning, or the electromagnetic waves generated by natural phenomena such as volcanoes and earthquakes.
KAIDI level transmitter manufacturer is dedicated in providing complete customized solutions for a wide range of industrial automation process applications – in material level, liquid flow, pressure and temperature. We are constantly developing and innovating, our core vision – “to provide solutions that exceed customers’ expectations. In 2012, the company successfully expanded its operations both locally and internationally, achieving global success and recognition for quality fork type level switch, magnetic level gauge products and services. Discover more details on kaidi86.com. Suitable for chemical industry, petroleum industry, metallurgical industry, water conservancy and electronic industry, etc.
Product features: Large display range, wide range of use, suitable for low temperature, medium temperature, high pressure occasions. It can be matched with remote transmitter output 4-20MADC standard signal to realize remote display of liquid level guage. It can also be matched with liquid level switch to realize liquid level control. Generally speaking, the material level feedback is inaccurate and untimely, which is very prone to explosion events, and there is a greater safety risk. Therefore, the requirements for the radar level meter are very high, but the radar level meter often has abnormal material level jumps or falls, or even crashes, resulting in large errors in material level measurement, which brings trouble to production and sales.